Licel Ethernet Connectivity
The new family of ethernet control modules provide a easy way for detection system remote control over the Internet
The communication with the ethernet interface uses TCP/IP . The controller can have a static IP address or a dynamically assigned address (DHCP). The controller accepts commands with ASCII syntax similar to the GPIB syntax. The twisted pair connector makes integration into the laboratory network easy. The different modules can be stacked together so that only one network connection is necessary.
Transient Recorder Remote Control
- Control and readout of up to 16 transient recorders
- LabView driver similar to the DIO-32HS driver
- Efficient Push-Mode for single shot acquisitions
- Part# Ethernet-int
Detector Remote Control
The detector remote control is offered in different configurations: for 1, 4 or 8 PMT's or for 4 APD's
PMT Remote Control
The PMT remote control set and reads back the PMT high voltage. PMT's with a activated high voltage are indicated trough a LED.
The PM-Remote-4 goes perfectly with a Rack-2.
The PM-Remote-8 is the optimum choice for a PMT-Rack8.
APD Remote Control
With a APD remote control (APD-Remote-4) unit the APD high voltage can be set and the temperature controller can be turned on and off
The Licel Trigger Module (Pretrig)incorporates one trigger input and 4 different outputs to build up compact detection
systems. The triggger input can be used to synchronize the system to an external laser flash lamp or Q-switch
trigger. The module can also run internally triggered. The module is able to generate:
All timings are derived from a quarz based oszillator ensuring nanosecond timing stability.
- a lamp trigger
- a pretrigger for the transient recorder
- a Q-Switch trigger
- and a Gating trigger for gated PMT-Modules.
The programming manual gives a complete description of the setup procedure and the command syntax. All modules come together with a LabView source code and a sample application.
Back to index